Display: multipositional Moving 19” LCD
Control Second Display: 10.4” touch screen monitor
Power: (AC) 220V±22V 50Hz±1Hz
Input power: 350VA
Protection against harmful ingress of water?
Protection against electric shock : Class I.
Degree of protection against electric shock: Type BF.
IPX1 the probe cable?IPX1
IPX7 the traducer of the probe?IPX7
Scope of application
- General and Special (Vascular)
- Obstetrics and Gynecology
- Musculoskeletal system
- Vascular and Peripheral vessels
- Small parts (including breasts, thyroid, testis, etc.)
Electronic linear, electronic convex, Electronic phased array
Mode of display
Black-and-white image: B, 2B, 4B, right/left B|M, B|D, right/left PW(D), top/down PW(D), M, B mode local amplification, linear probe trapezoidal imaging, 3D reconstruction
Blood flow images: spectrum: B/BC, BC/B, B/D, B/C/D CFM: B|C|D, B|C|M, B|C double real-time, PW(D), CFM, CPA, Power doppler
Color Doppler: color mapping, Tissue harmonic imaging, image enhancing algorithm.
Mode of display: energy, velocity, variance (in support of velocity variance), frequency, gain
Display control: the size and position of the sampling box are adjustable, the emitting angle of the linear probe is adjustable.
Sampling volume and position are adjustable
PW(D) tracing supports maximum and average velocity. The threshold of the maximum velocity is adjustable. Scanning scope: above base-line, below base-line and entire scanning.
PW(D) sound can be turned on or off, and can be regulated in volume. Velocity measuring: in support of maximum and minimum velocity
Parameters for control of blood flow color images: Doppler frequency, position and size of sampling box, base line, color gain, deflection angle, wall filter, accumulative number of times, etc.
Color depth of display
Electronic focusing + sound lens focusing
Emission involves single focus, double focuses, triple focuses, and quadruple focuses
Continuously dynamic focusing in receiving
Formation of digital beam
Continuously dynamic focusing
Dynamic aperture diameter
Weighted sum of reception delay
In support of semi-moment scanning, and ±10°linear reception deflection angle
Signal processing/Doppler sound
Dynamic wave filtering
Total gain control
Composite gain technology: 8-section TGC and D-AGC.
Acoustic output adjustment
Adjustable Doppler stereophonic acoustic output
Dynamic scope variation and logarithmic compaction
Gray scale variation
B|M- or M-mode scanning speed
Linear density control
Scanning angle/width control
Upward and downward Image flipping
Multi-angle image rotation
Options for color gray-scale bar inversion
Combined processing of tissue and flow images
Resolution of black-and-white images: 1024×768×8 bits (256 gray-scale)
Resolution of color images: 8 bit×8 bit×8 bit color coding
Selection of specific images
Optimized items are available to choose from
Basic measuring and calculating functions
2D normal measurement: distances, angle, perimeter, area (ellipse and locus methods), volume, volume – 3-line method (length, width, height), histogram, cross-section chart
Basic M-Mode measurement: time, distance, heart rate, valve rate
With 4 varieties of obstetric databases for reckoning gestation age
3 non-editable versions: Asian, European and American, plus 1 editable version: user-customized.
The gestation age can be reckoned by the information in each version on gestational sac diameter, crown-rump length, biparietal diameter, head circumference, abdominal circumference, femur length, humerus length, abdominal transversal diameter, length of vertebrae, occipital-frontal diameter, cerebellum diameter, fibula length, radius length, Cisterna Magna, inner orbital diameter, output orbital diameter, amniotic fluid index, fetal heart rate, nuchal transparency, distance and area, etc.
Measurement and calculation of amniotic fluid index (AFI)
Ratio calculation (BPD/OFD, FL/AC, FL/BPD and HC/AC)
Fetus weight estimation
Reckoning the gestation age and EDD by LMP and BBT
Fetal biophysical score
Fetal growth curve
Andrology measurement and calculation
Measurement and calculation of prostate and testis
Calculation of predicted prostatic specific antigen (PPSA) and Prostatic Specific Antigen density (PSAD).
Measurement and calculation of the uterus, left ovarium, right ovarium, left folliculus, right folliculus, etc.
Urology measurement and calculation
Detection and calculation of the left kidney, right kidney, bladder, residual urinal vol, etc.
Vessel measurement and calculation
Measurement and calculation of area stenosis rate and vascular stenosis rate.
Multi-fetus detection and measurement
Detection of multiple fetuses
Small parts detection and calculation
Detection and calculation of thyroid, galactophore, masses, etc.
Cardiac detection and calculation functions
Cardiology measurement software package—contains features of analyzing and measuring aorta, mitral valve, (left/right) ventricles, etc.
Area stenosis percentage (% area sten), tube diameter stenosis percentage (% Diam Sten)
Body surface area (BSA)
In terms of bodily function, an body mark database is divided into several parts: abdomen, andrology, cardiology, gynecology, musculoskeletal, obstetrics,
pediatric, small parts, urology, etc.
Probe conditions, mode of display, depth, focus, dynamic range, body marks, probe position marker, acoustic output, patient conditions, name of medical agency, measured value, time and date, scale, scanning direction, gray-scale curve, current working frequency of the probe, frame frequency, B-Mode total gains, C-Mode total gains, Mode-D total gains, menu, annotations, gray-scale belt, puncture guide line, TI (thermal index), MI (mechanical index)
Probe parameter storage
Cine loop storage
Measurement results storage
Cine-loop speed setting
Frame-by-frame forward/backward cine-loop
Chinese and English available
Function of the trackball
Performing various operations based on the working state and menu
Controlling multiple electronic scale, locating the cursor, completing measurement
Input/output port Achieving: USB. BW printer, CR/RW
VGA output port
DVI output port
Video port (for B/W printer)
Ultrasound Probe technical specifications
Physical Number of channels 128
3.5MHzR65 multi-frequency convex probe (128 channels) , 192 crystals
The probes have a frequency range between 2.0MHz-5.5MHz. In the B-Mode, have a scanning angle of 60°. In the M-Mode, it can activate a particular array element as fixed-frequency probes. It can be used in the combination of B-Mode and M-Mode.
Scope of application: examination and diagnosis of abdominal tissues (liver, gallbladder, spleen, pancreas, kidneys, gastro-intestinal, uterus, urinary bladder, prostate, etc.), assessment of growth and maturity of the fetus in obstetric diagnosis.
Endocavity probes (transvaginal and transrectal)
6.5MHzR10 multi-frequency convex probe (128 channels) , 192 crystals
The probes have a frequency range between 5.0MHz-9.0MHz. In the B-Mode , they have a scanning angle of 140-158°. In the M-Mode, they can activate particular array element as fixed-frequency probes. They can be used in the combination of B-Mode and M-Mode.
Scope of application: vagina, uterus, rectum, rectum wall, prostate, etc.
Warning: A sterile rubber sleeve must be used with this probe. The probe should not be activated before examination.
7.5MHzL46 multi-frequency linear probe (128 channels), 192 crystals
Frequency range between 6.0MHz-12.0MHz. In the B-Mode, has a scanning width of 38-45°. In the M-Mode, it can activate a particular array element as a fixed-frequency probe. Can be used in the combination of B-Mode and M-Mode.
7.5MHz multi-frequency rectal probe (128 channels), 192 crystals
Frequency range between 6.5MHz-8.5MHz. In the M-Mode, can activate a particular array element as a fixed-frequency probe. Can be used in the combination of B-Mode and M-Mode.
Scope of application: Rectum
Capability to add different options eg:
DICOM, Stress Echo, Elastography, can also be added after purchase.